|Place of Origin:||China|
|Certification:||CNAS Calibration Certificate|
|Minimum Order Quantity:||1 piece|
|Packaging Details:||Each in a black gift box then into a carton box|
|Delivery Time:||15 work days|
|Payment Terms:||Western Union, T/T, MoneyGram|
|Service:||OEM/ODM||Figure No.:||7006-25-7, 7006-26-4, 7006-30-2, 7006-30A-1, 7006-31-4|
Calibrated High Quality Steel IEC60061-3 Go Gauges for Simple Pass Test or Not Pass Test of E40 Lamp Cap
Lamp cap and lampholder go no go gauges are designed according to IEC60061-3:2004, they are applied to quality test of E40 Lamp cap. The structure and waveform are calculated by PC to reach high precision, and the appearance looks nice. The material is applied alloy steel. All the products equip with third-lab certificate authorized by CNAS.
|E40 Lamp Cap gauge|
|7006-27-7||“Go” gauge for caps on finished lamps|
|7006-28D-1||“Not Go” Gauge for caps on finished lamps|
|7006-52-1||Gauge for finished lamps fitted with E40 caps for testing contact making|
|7006-53-1||Gauge for finished lamps fitted with E40 caps for testing protection against accidental contact|
E40 Lamp Cap Go No Go Gauge
1) “Go” gauge for caps on finished lamps (7006-27-7)
Purpose: To check the maximum dimensions of the screw thread and dimension T1 minimum of E40 caps on finished lamps.
Testing: When the cap in a finished lamp has been screwed into the gauge as far as it will go, the centre contact shall be co-planar with, or project beyond, surface X.
2)“Not Go” Gauge for caps on finished lamps (7006-28D-1)
Purpose: To check the minimum outside (major) diameter of the screw thread, dimension d, of E40 caps on finished lamps.
Testing: When the gauge is placed over the thread of the cap on a finished lamp, held cap uppermost, the centre contact shall not project beyond surface X.
Only the weight of the gauge itself shall be used in the test.
3)Gauge for finished lamps fitted with E40 caps for testing contact making (7006-52-1)
Purpose: To check lamp dimensions for contact making in I.E.C. Holders.
Testing: The shape of the lamp with regard to the fit in the lampholder is assumed to be correct. If the lamp can be pushed into the gauge till plane W reached plane V or projects beyond it.
4)Gauge for finished lamps fitted with E40 caps for testing protection against accidental contact (7006-53-1)
Purpose: To check protection against accidental contact.
Testing: The shape of the lamp with regard to protection against accidental contact is assumed to be correct if the test sample being pushed as far as possible into the gauge, plane W does not protrude beyond plane V.
|E40 Lampholder Gauge|
|7006-25-7||Go Gauge for screw threads of lampholder|
|7006-26-4||Not Go Gauge for screw threads of lamphoder|
|7006-23-3||Gauge for protecting against Bulb-neck damage and for testing contact-making in lampholder|
|7006-24-3||Plug gauge for lampholders for testing protection against accidental contact|
|BS EN60238 Figure 5||Test Cap for Lampholders E40|
|BS EN60238 Figure 11||Test Cap for lampholders E40|
|BS EN60238 Figure 15||Test Cap for lampholders E40|