|Place of Origin:||China|
|Certification:||CNAS Calibration Certificate|
|Minimum Order Quantity:||1 piece|
|Packaging Details:||Each in a black gift box then into a carton box|
|Delivery Time:||15 work days|
|Payment Terms:||Western Union, T/T, MoneyGram|
|Service:||OEM/ODM||Figure No.:||7006-27D-3, 7006-29L-4, 7006-29A-2, 7006-29-3, 7006-29C-2, 7006-29A-2|
Calibrated High Quality Steel IEC60061-3 Go Not Go Gauges for Simple Pass Test or Not Pass Test of E26 Screw Cap
Lamp cap and lampholder go no go gauges are designed according to IEC60061-3:2004, they are applied to quality test of E26 lamp screw cap. The structure and waveform are calculated by PC to reach high precision, and the appearance looks nice. The material is applied alloy steel. All the products equip with third-lab certificate authorized by CNAS.
|E26 lamp cap gauge|
|7006-27D-3||“Go” gauge for caps|
|7006-29L-4||“Not Go” gauge for caps|
|7006-29-3||Gauge for testing contact-making of lamps fitted with caps|
1)“Go” Gauge for Caps E26, E26/50*39 and E26d (7006-27D-3)
Purpose: to check the maximum dimensions of the screw thread and dimension T1 min of E26, E26/50*39, E26/51*39 or E26d caps on finished lamps.
Testing: when the cap on a finished lamp has been screwed into the gauge as far as it will go, the central contact shall touch surface X. When the lamp is removed from the gauge, at least two full turns shall be required disengage the thread.
2)“Not Go” Gauge for Caps E26, E26/50*39, E26/51*39 and E26d (7006-29L-4)
Purpose: to check minimum outside (major) diameter of the screw thread, dimension d, of E26, E26/51*39 or E26d caps on finished lamps.
Testing: when the gauge is placed over the thread of the cap on a finished lamp, held cap uppermost, the central contact shall not project beyond surface X.
3)Gauge for Caps E26d on Finished Lamps for Testing Protection Against Accidental Contact (7006-29A-2)
Purpose: to check protection against accidental contact of caps E26d on finished lamps.
Testing: when the lamp has been pushed into the gauge as far as it will go with the side solder of the cap located into recess provided, surface W of the plunger shall not project beyond surface V.
4)Gauge for Testing Contact-Making of Lamps Fitted with Caps E26, E26/50*39, E26/51*39 (7006-29-3)
Purpose: to check lamp dimensions for contact-making in E26, E26/50*39, E26/51*39 or E26d lampholders.
Testing: the shape of the lamp with regard to the fit in the lampholder is assumed to be correct, if the lamp can be pushed into the gauge until plane W reaches plane V or projects beyond it.
For E26/24 capped lamps, when surfaces V and W are coincident, the top of the cap including any solder or conductive material, shall not above surface X.
5)Gauge with Reference Diameter of 13.2mm for Caps E26d (7006-29C-2)
Purpose: to check dimensions C max and C1 max of caps E26d.
Testing: when the unmounted cap has been pushed into the gauge as far as it will go, surface W of the plunger shall not project beyond surface X. For caps on finished lamps, surface W shall not project beyond surface V. In this instance, the cap shall be so positioned that the solder on the intermediate contact is located in the recess provided in the gauge.