|Place of Origin:||China|
|Certification:||CNAS Calibration Certificate|
|Minimum Order Quantity:||1 piece|
|Packaging Details:||Each in a black gift box then into a carton box|
|Delivery Time:||15 work days|
|Payment Terms:||Western Union, T/T, MoneyGram|
|Service:||OEM/ODM||Figure No.:||7006-10-8, 7006-11-8, 7006-4A-2, 7006-3-1, 7006-19-2|
Calibrated High Quality Steel IEC60061-3 Not Go Gauges for Simple Pass Test or Not Pass Test of B22 Cap 7006-10-8
Lamp cap and lampholder go no go gauges are designed according to IEC60061-3:2004, they are applied to quality test of B22 lamp cap. The structure and waveform are calculated by PC to reach high precision, and the appearance looks nice. The material is applied alloy steel. All the products equip with third-lab certificate authorized by CNAS.
|B22 Lamp Cap Gauge|
|7006-10-8||Not Go Gauge for Caps on Finished Lamps|
|7006-11-8||Go gauge for Caps on Finished lamps BA9, B15, BA15 & B22|
|7006-4A-2||Gauges for Testing the Insertion of Caps in lampholders B15d, B22d & BY22d|
|7006-4B-1||Gauges for Testing the Retention of B15d and B22d Caps in the Holder|
|7006-3-1||Acceptance Gauge for B22d Caps Intended for Automatic Wire Threading|
|7006-17A-1||Go Gauge for Cap on Finished Lamps BY22d|
1)“Not Go” Gauge for B22 Caps on Finished Lamps (7006-10-8)
Purpose: to check dimension A min. of caps B22 on finished lamps.
Testing: caps on finished lamps shall be assured to be correct if the gauge does not pass over the cap by its own weight.
2)“Go” Gauges for B22 Caps on Finished Lamps(7006-11-8)
Purpose: to check dimension A max, N min, D1 max, and the diametrial position of the pins of caps B22 on finished lamps.
Testing: the cap shall enter the gauge until the pins have passed through the slots Q. The cap is then turned through a small angle and is pressed so that the pins are in close contact with surface S. In this position, the contact-making surface shall not be below surface X nor shall it project beyond surface Z.
3)Gauges for Testing the insertion of Caps in B22 Lampholders (7006-4A-2)
Purpose: to check B22d cap dimensions for insertion of the cap on the finished lamp in the lampholder.
Testing: without using undue force it shall be possible to insert the cap into the gauge until the pins have passed completely through the slots.
4)Acceptance Gauge for B22d Caps intended for Automatic Wire Threading (7006-3-1)
Purpose: to check the suitability for automatic wire threading of B22d caps as shown on sheet 7004-10.
Testing: the cap be assumed to be correct if it can be inserted smoothly into the gauge until the contact plates of the cap touch surface Y of the gauge
5)“Go” Gauge for Cap B22d -3(900/1350)/25*26 on Finished Lamps(7006-19-2)
Purpose: for the control of dimensions A max, N min, D min, D1 max, and the angular position of the pins illustrated on sheet 7004-10A.
Testing: the cap shall enter the gauge until the pins have passed through the slots Q. The cap is then turned through a small angle and is pressed so that the pins are in close contact with surface S. In this position the contact making surface shall not be below surface X nor shall it project beyond the surface Z. “N” of the gauge tests diameter of the cap for a sufficient length to ensure interchangeability of the caps in the holder.