|Place of Origin:||China|
|Certification:||CNAS Calibration Certificate|
|Minimum Order Quantity:||1 piece|
|Packaging Details:||Each in a black gift box then into a carton box|
|Delivery Time:||15 work days|
|Payment Terms:||Western Union, T/T, MoneyGram|
|Standard:||IEC60432-1 And IEC60560-2||Type:||B15|
Calibrated IEC60432-1 Figure C.1 and IEC60560-2 Figure 4 Torque Gauges for Torsion Resistance of B15 Unused Lamps
B15 Torque Gauge is met the requirements of IEC60432-1 figure C.1 and IEC60560-2 figure 4
|Dimension||B15 (mm)||B22 (mm)||Tolerance (mm)|
Relative Gauges of IEC60061-3 B15 lamp Cap and holder Gauge
|B15 Lamp Cap Gauge|
|7006-4A-2||B15 Lamp Cap and holder GO and Not Go Gauges|
|7006-4B-1||Gauges for testing the retention of B15d and B22d caps in the holder|
|7006-10-8||Not Go gauges for caps on finished lamps BA9, B15, B22 and BY22d|
|7006-11-8||Go gauges for caps on finished lamps BA9, B15, BA15 and B22|
|B15 Lampholder Gauge|
|7006-12C-2||1)Supplementary “Go” Gauge for lampholders B15|
|7006-12D-2||“Not Go”/Retention Gauge for Lampholders B15|
|7006-15B-1||Gauges for Checking Contact Position and Contact Force in Bayonet Lampholders B15d|
1)Gauges for testing the insertion of caps in lamphoder B15d, B22d and BY22d (Figure No.: 7006-4A-2)
Purpose: to check B15d, B22d and BY22d cap dimensions for insertion of the cap on finished lamp in the lampholder.
2)Gauges for testing the retention of B15d and B22d caps in the holder (7006-4B-1)
Purpose: to check cap dimension for retention of the cap on the finished lamp in the lampholder.
3)Not Go gauges for caps on finished lamps BA9, B15, B22 and BY22d (7006-10-8)
Purpose: to check dimension A min. of caps BA9, B15, B22 and BY22d on finished lamps respectively.
4)Go gauges for caps on finished lamps BA9, B15, BA15 and B22 (7006-11-8)
Purpose: to check dimensions A max, N min, D1 min, D1 max, and the diametrical positions of the pins of caps BA9, B15, BA15 and B22 on finished lamps.
B15 Lampholder Gauge
1)Supplementary “Go” Gauge for lampholders B15 (7006-12C-2)
Purpose: The gauge shall be used solely to check interchangeability with respect to the corresponding caps on finished lamps after the test of clause 18: General resistance to heat, of IEC61184: Bayonet lampholders.
Testing: It shall be possible to inset the gauge into the holder and to locate the pins at the resting points without using undue force.
Purpose: The use of this gauge is restricted to the following purposes:
In this position it shall not be possible to remove the gauge when an axial pull is applied to it. The test is repeated at the other “J” slot.
4)Gauges for Checking Contact Position and Contact Force in Bayonet Lampholders B15d (7006-15B-1)
Purpose: To check lampholders B15d with respect to: